This lancing device is still supported by Roche Diabetes Care however the device is no longer for sale.
The Accu-Chek Multiclix is designed with a built-in 6 lancet exchange system. No more accidental finger-stick injuries. Accu-Chek Multiclix makes testing gentle, easy and virtually pain free1.Product Manuals
1. Kocher, S, et al., Comparison of Lancing Devices for Self-Monitoring of Blood Glucose Regarding Lancing Pain J Diabetes Sci Technol 2009; Sep. 3(5):1136-1143.
Control penetration with 11 depth settings
0.7 to 2.2 mm
Prime button to prime and yellow release button
Yes, the number of available lancets is indicated in the window.
A blue ring and red mark is displayed on a used lancet drum.
No, not without misuse.
Always use a new lancet when you obtain blood. In doing this, you reduce the risk of infection, and obtaining blood remains virtually pain-free.
0.3 mm (30G)
Always wash and dry hands before testing.
Inserting a lancet drum
Setting the penetration depth
Turn the rotatable cap to select from 11 settings. Start with 2, and adjust depending on your skin type.
Obtaining a blood sample
Press the priming button to prime. Press the yellow button to release the lancet.
Loading a new lancet
Twist the priming button to change to the next lancet. The numbers on the window show how many lancets are left.